Abstract of Pill Camera
The aim of technology is to make products in a large scale for cheaper prices and increased quality. The current technologies have attained a part of it, but the manufacturing technology is at macro level. The future lies in manufacturing product right from the molecular level. Research in this direction started way back in eighties. At that time manufacturing at molecular and atomic level was laughed about. But due to advent of nanotecnlogy we have realized it to a certain level. One such product manufactured is PILL CAMERA, which is used for the treatment of cancer, ulcer and anemia. It has made revolution in the field of medicine. This tiny capsule can pass through our body, without causing any harm.
Description :Pill Camera
The device, called the given Diagnostic Imaging System, comes in capsule form and contains a camera, lights, transmitter and batteries. The capsule has a clear end that allows the camera to view the lining of the small intestine. Capsule endoscopy consists of a disposable video camera encapsulated into a pill like form that is swallowed with water. The wireless camera takes thousands of high-quality digital images within the body as it passes through the entire length of the small intestine. The latest pill camera is sized at 26*11 mm and is capable of transmitting 50,000 color images during its traversal through the digestive system of patient.
Video chip consists of the IC CMOS image sensor which is used to take pictures of intestine .The lamp is used for proper illumination in the intestine for taking photos. Micro actuator acts as memory to store the software code that is the instructions. The antenna is used to transmit the images to the receiver. For the detection of reliable and correct information, capsule should be able to designed to transmit several biomedical signals, such as pH, temp and pressure. This is achieved with the help of Soc.external reference crystal or clock. The decoder IC receives the serial stream and interprets the serial information as 4 bits of binary data. Each bit is used for channel recognition of the control signal from outside the body. Since the CMOS image sensor module consumes most of the power compared to the other components in the telemetry module, controlling the ON/OFF of the CMOS image sensor is very important.
Moreover, since lightning LED’s also use significant amount of power, the individual ON/OFF control of each LED is equally necessary. As such the control system is divided into 4 channels in the current study. A high output current amplifier with a single supply is utilized to drive loads in capsule.